According to FDA Recall Notice, Medtronic is recalling the SynchroMed Implantable Infusion Pumps because a software problem may cause unintended delivery of drugs during a priming bolus procedure, used to quickly deliver large dose of medication from the device to the patient’s spine. During this procedure, patients may receive the drug unintentionally at a high rate of infusion in the cerebrospinal fluid followed by a period of reduced drug delivery after the priming bolus. This can result in a drug overdose or under dose which can lead to serious adverse health consequences such as respiratory depression, coma or death.
The SynchroMed II and SynchroMed EL Implantable Drug Infusion Pumps (SynchroMed Implantable Infusion Pumps) are programed to deliver prescribed drugs to a specific site inside the patient’s body. The SynchroMed pumps are used to treat primary or metastatic cancer, chronic pain, and severe spasticity.